Mission History of Reflective Solar Band Calibration Performance of VIIRS
Environmental Data Records (EDR) from the Visible Infrared Imaging Radiometer Suite (VIIRS) have a need for Reflective Solar Band (RSB) calibration errors of less than 0.1%. Throughout the mission history of VIIRS, the overall instrument calibrated response scale factor (F factor) has been calculated with a manual process that uses data at least one week old and up to two weeks old until a new calibration Look Up Table (LUT) is put into operation. This one to two week lag routinely adds more than 0.1% calibration error. In this paper, we discuss trending the solar diffuser degradation (H factor), a key component of the F factor, improving H factor accuracy with improved bidirectional reflectance distribution function (BRDF) and attenuation screen LUTs , trending F factor, and how using RSB Automated Calibration (RSBAutoCal) will eliminate the lag and look-ahead extrapolation error.